Profile name or number (9 characters)
Number of steps in the test. Parameter can differ for each step.
Ref. Pos. Set. - Reference Position Set
Sets the starting position of the cross-head by moving the cross-head manual to the desired position with the dial-knob. Press on the dial-knob to put the cross-head in motion.
Ref. Pos. - Reference Position
Starting position of the test. Once a test profile has been selected the test automatically travels to the reference point. (in, mm)
All steps of the test will run for the specified numbers of cycles.
Cross-head automatically returns to the reference point after a peak has been detected or the test has been completed.
When enabled the test stand stops if a sample break is detected. The test stand performs its own break (peak) calculation from the measurements it receives from the force gauge.
Starting direction of the cross-head from the reference position.
Time: The amount of time the cross-head paused.
The cross-head maintains specified constant force. The cross head performs minimum movements required to maintain a constant force applied within a specified tolerance range for a specified amount of time.
Duration: The amount of time required to maintain the specified constant force.
Force: The amount of force which will be maintained for a specified time.
Tolerance: Force tolerance specified in percent +/- used to maintain the position of the cross-head in order to maintain the specifies force.
Moving speed of the cross-head (in/min, mm/min)
Distance: The distance required for the cross-head to travel.
Duration: Step will run for a specified amount of time.
Step stops if a sample break is detected. The test stand performs its own break (peak) calculation from the measurements it receives from the force gauge.
Step ends when a set force conditions is met.
Condition: Greater than or less than condition for the force value at which the step will end.
Force: Specified force condition value
Step ends when cross-head reached the set position.